Right- click on My Computer, select Properties, Here are some ways to work around drivers that are not behaving properly: Figure Replace the cover plate screw Connecting instruments to the B Installation and Configuration Guide 19 1 Installing and Configuring the After the is installed in your PC, the next step is to connect your GPIB instruments to the installed Agilent Technologies makes no warranties of any kind with regard to this document, whether express or implied. Since Device Manager may have disabled the device, click Enable to restart the
|Date Added:||13 November 2007|
|File Size:||55.41 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Agilent Technologies makes no warranties of any kind with regard to this document, whether express or implied. Allow Windows to find and install the driver automatically.
If several devices exist on a bus, be sure each has a unique GPIB bus address and only one device is the System Controller. If doing Steps 1, 2, 3 or 4 does not solve the problem, reboot the PC.
The interface is supported on Windows NT 4. Product price, specification, warranty and availability subject to change without notice. The length between adjacent devices is not critical as long as the overall restriction is met. Accessing Programming Manuals and Help You can access. If none work, you will have to free up other IRQs by disabling devices such as serial ports or sound cards.
Some additional factors to consider are: The GPIB connector screws should be finger- tightened only. To get started, you may want to copy the figure to use as a guide as you set up your GPIB system.
Agilent 82350B PCI GPIB Interface Installation and
An improperly attached GPIB connector can cause the bus to malfunction. Please help us maintain a helpfull driver collection.
Microsoft Internet Explorer 5. Caution, risk of danger refer to this manual for specific Warning or Caustion information.
Agilent Technologies B User Manual – Page 1 of 69 |
This section shows one way to agilet Interactive IO to verify instrument communication. The Web version of the manual will always be the latest revision. See the associated steps if you need more details.
If you choose not to reboot at this time, you should reboot before running Connection Expert. If none work, you will have to free up other IRQs by disabling devices such as serial ports or sound cards. Clicking the blue IO icon and then clicking Exit causes a dialog box Further, to the maximum extent permitted by applicable law, Agilent disclaims all warranties, either express or implied, with regard to this manual and any information contained herein, including but not limited to the implied warranties of merchantability and fitness for a particular purpose.
The auto- discovery process can cause sound and other cards to stop responding.
Agilent B PCI GPIB Interface_百度文库
Handle the card ONLY by the sheet metal frame or by the card edges. In general, any x86 or x64 except Itanium should work but there may be a decrease in performance. New computers 2 3 oftentimes have newer BIOSs available. Documents Flashcards Grammar checker. Agilent Technologies b Pci Gpib driver installation manager was reported as very satisfying by a large percentage of our reporters, so it is recommended to download and install.
Replace cables as required.
In this case, or if the installed version is technologirs L. These addresses are chosen by convention but any address in the range 0 – 30, inclusive, may be used. Installing and Configuring the 1 14 After the files have been copied, you may see a dialog asking System Controller Determines if this interface controls which bus devices talk and which bus devices listen.
Hiding Agilent IO Control. Technolgies Symbols The following symbol on the instrument and in the documenta- tion indicates precautions that must be taken to